Infobox chemical analysis
name = Gas-liquid chromatography
caption =A gas chromatograph with a headspace sampler
acronym = GLC, GC
analytes = organic
must be volatile
Agilent(a spin-off of Hewlett-Packard) PerkinElmer Shimadzu Thermo Electron Corporation Varian, Inc.
Thin layer chromatography High performance liquid chromatography
Gas chromatography-mass spectrometry
Gas-liquid chromatography (GLC), or simply gas chromatography (GC), is a type of
chromatographyin which the mobile phase is a carrier gas, usually an inertgas such as heliumor an unreactive gas such as nitrogen, and the stationary phase is a microscopic layer of liquid or polymeron an inert solid support, inside glass or metal tubing, called a column. The instrument used to perform gas chromatographic separations is called a "gas chromatograph" (also: "aerograph", "gas separator").
Gas Chromatography is different from other forms of chromatography (
HPLC, TLC, etc.) because the solutions travel through the column in a gas state. The interactions of these gaseous analytes with the walls of the column (coated by different stationary phases) causes different compounds to elute at different times called retention time. The comparison of these retention times is the analytical power of GC. This makes it very similar to HPLC.
Chromatographydates to 1903 in the work of the Russian scientist, Mikhail Semenovich Tswett. German graduate student Fritz Priordeveloped solid state gas chromatography in 1947. Archer John Porter Martin, who was awarded the Nobel Prizefor his work in developing liquid-liquid (1941) and paper (1944) chromatography, laid the foundation for the development of gas chromatography and later produced liquid-gas chromatography (1950).
A gas chromatograph is a chemical analysis instrument for separating
chemicals in a complex sample. A gas chromatograph uses a flow-through narrow tube known as the "column", through which different chemical constituents of a sample pass in a gas stream (carrier gas, "mobile phase") at different rates depending on their various chemical and physical properties and their interaction with a specific column filling, called the "stationary phase". As the chemicals exit the end of the column, they are detected and identified electronically. The function of the stationary phase in the column is to separate different components, causing each one to exit the column at a different time ("retention time"). Other parameters that can be used to alter the order or time of retention are the carrier gas flow rate, and the temperature.
In a GC analysis, a known volume of gaseous or liquid
analyteis injected into the "entrance" (head) of the column, usually using a micro syringe(or, solid phase microextraction fibers, or a gas source switching system). As the carrier gas sweeps the analyte molecules through the column, this motion is inhibited by the adsorptionof the analyte molecules either onto the column walls or onto packing materials in the column. The rate at which the molecules progress along the column depends on the strength of adsorption, which in turn depends on the type of molecule and on the stationary phase materials. Since each type of molecule has a different rate of progression, the various components of the analyte mixture are separated as they progress along the column and reach the end of the column at different times (retention time). A detector is used to monitor the outlet stream from the column; thus, the time at which each component reaches the outlet and the amount of that component can be determined. Generally, substances are identified (qualitatively) by the order in which they emerge (elute) from the column and by the retention time of the analyte in the column.
The autosampler provides the means to introduce automatically a sample into the inlets. Manual insertion of the sample is possible but is no longer common. Automatic insertion provides better reproducibility and time-optimization.
Different kinds of autosamplers exist. Autosamplers can be classified in relation to sample capacity (auto-injectors VS autosamplers, where auto-injectors can work a small number of samples), to robotic technologies (XYZ robot VS rotating/SCARA-robot – the most common), or to analysis:
* Static head-space by syringe technology
* Dynamic head-space by transfer-line technology
Traditionally autosampler manufactures are different from GC manufactures and currently no GC manufacture offers a complete range of autosamplers. Historically, the countries most active in autosampler technology development are the United States, Italy, and Switzerland.
The column inlet (or "injector") provides the means to introduce a sample into a continuous flow of carrier gas. The inlet is a piece of hardware attached to the "column head".
Common inlet types are:
* S/SL (Split/Splitless) injector; a sample is introduced into a heated small chamber via a syringe through a "septum" - the heat facilitates
volatilizationof the sample and sample matrix. The carrier gas then either sweeps the entirety ("splitless" mode) or a portion ("split" mode) of the sample into the column. In split mode, a part of the sample/carrier gas mixture in the injection chamber is exhausted through the "split vent".
* On-column inlet; the sample is here introduced in its entirety without heat.
* PTV injector; Temperature-programmed sample introduction was first described by Vogt in 1979. Originally Vogt developed the technique as a method for the introduction of large sample volumes (up to 250 µL) in capillary GC. Vogt introduced the sample into the liner at a controlled injection rate. The temperature of the liner was chosen slightly below the boiling point of the solvent. The low-boiling solvent was continuously evaporated and vented through the split line. Based on this technique, Poy developed the Programmed Temperature Vaporising injector; PTV. By introducing the sample at a low initial liner temperature many of the disadvantages of the classic hot injection techniques could be circumvented.
* Gas source inlet or gas switching valve; gaseous samples in collection bottles are connected to what is most commonly a six-port "switching valve". The carrier gas flow is not interrupted while a sample can be expanded into a previously evacuated "sample loop". Upon switching, the contents of the sample loop are inserted into the carrier gas stream.
* P/T (Purge-and-Trap) system; An inert gas is bubbled through an aqueous sample causing insoluble volatile chemicals to be purged from the matrix. The volatiles are 'trapped' on an absorbent column (known as a trap or concentrator) at ambient temperature. The trap is then heated and the volatiles are directed into the carrier gas stream. Samples requiring preconcentration or purification can be introduced via such a system, usually hooked up to the S/SL port.
* SPME (solid phase microextraction) offers a convenient, low-cost alternative to P/T systems with the versatility of a syringe and simple use of the S/SL port.
Two types of columns are used in GC:
*Packed columns are 1.5 - 10 m in length and have an internal diameter of 2 - 4 mm. The tubing is usually made of stainless steel or glass and contains a "packing" of finely divided, inert, solid support material (eg.
diatomaceous earth) that is coated with a liquid or solid stationary phase. The nature of the coating material determines what type of materials will be most strongly adsorbed. Thus numerous columns are available that are designed to separate specific types of compounds.
*Capillary columns have a very small internal diameter, on the order of a few tenths of millimeters, and lengths between 25-60 meters are common. The inner column walls are coated with the active materials (WCOT columns), some columns are quasi solid filled with many parallel micropores (PLOT columns). Most capillary columns are made of fused-
silicawith a polyimideouter coating. These columns are flexible, so a very long column can be wound into a small coil.
*New developments are sought where stationary phase incompatibilities lead to geometric solutions of parallel columns within one column. Among these new developments are:
** Internally heated "microFAST" columns, where two columns, an internal heating wire and a temperature sensor are combined within a common column sheath ( [http://etd.lsu.edu/docs/available/etd-04012004-193144 microFAST] );
** " [http://www.restek.com/restek/images/external/620-04-001.pdf Micropacked columns] " (1/16" OD) are column-in-column packed columns where the outer column space has a packing different from the inner column space, thus providing the separation behaviour of two columns in one. They can be easily fit to inlets and detectors of a capillary column instrument.
The temperature-dependence of molecular adsorption and of the rate of progression along the column necessitates a careful control of the column
temperatureto within a few tenths of a degree for precise work. Reducing the temperature produces the greatest level of separation, but can result in very long elution times. For some cases temperature is ramped either continuously or in steps to provide the desired separation. This is referred to as a temperature program. Electronic pressure control can also be used to modify flow rate during the analysis, aiding in faster run times while keeping acceptable levels of separation.
The choice of carrier gas ("mobile phase") is important, with hydrogen being the most efficient and providing the best separation. However, helium has a larger range of flowrates that are comparable to hydrogen in efficiency, with the added advantage that helium is non-flammable, and works with a greater number of detectors. Therefore, helium is the most common carrier gas used.
A number of detectors are used in gas chromatography. The most common are the
flame ionization detector(FID) and the thermal conductivity detector(TCD). Both are sensitive to a wide range of components, and both work over a wide range of concentrations. While TCDs are essentially universal and can be used to detect any component other than the carrier gas (as long as their thermal conductivities are different than that of the carrier gas, at detector temperature), FIDs are sensitive primarily to hydrocarbons, and are more sensitive to them than TCD. However, an FID cannot detect water. Both detectors are also quite robust. Since TCD is non-destructive, it can be operated in-series before an FID (destructive), thus providing complementary detection of the same analytes.
Other detectors are sensitive only to specific types of substances, or work well only in narrower ranges of concentrations. They include:
discharge ionization detector(DID)
electron capture detector(ECD)
flame photometric detector(FPD)
Hall electrolytic conductivity detector(ElCD)
helium ionization detector(HID)
nitrogen phosphorus detector(NPD)
mass selective detector(MSD)
pulsed discharge ionization detector(PDD)
thermal energy analyzer(TEA)
Some gas chromatographs are connected to a
mass spectrometerwhich acts as the detector. The combination is known as GC-MS. Some GC-MSare connected to an NMR spectrometer which acts as a back up detector. This combination is known as GC-MS-NMR. Some GC-MS-NMRare connected to an infrared spectrophotometer which acts as a back up detector. This combination is known as GC-MS-NMR-IR. It must, however, be stressed this is very rare as most analyses needed can be concluded via purely GC-MS.
The method is the collection of conditions in which the GC operates for a given analysis. Method development is the process of determining what conditions are adequate and/or ideal for the analysis required.
Conditions which can be varied to accommodate a required analysis include inlet temperature, detector temperature, column temperature and temperature program, carrier gas and carrier gas flow rates, the column's stationary phase, diameter and length, inlet type and flow rates, sample size and injection technique. Depending on the detector(s) (see below) installed on the GC, there may be a number of detector conditions that can also be varied. Some GCs also include valves which can change the route of sample and carrier flow, and the timing of the turning of these valves can be important to method development.
Carrier gas selection and flow rates
Typical carrier gases include
helium, nitrogen, argon, hydrogenand air. Which gas to use is usually determined by the detector being used, for example, a DID requires helium as the carrier gas. When analyzing gas samples, however, the carrier is sometimes selected based on the sample's matrix, for example, when analyzing a mixture in argon, an argon carrier is preferred, because the argon in the sample does not show up on the chromatogram. Safety and availability can also influence carrier selection, for example, hydrogen is flammable, and high-purity helium can be difficult to obtain in some areas of the world. (See: Helium--occurrence and production.)
The purity of the carrier gas is also frequently determined by the detector, though the level of sensitivity needed can also play a significant role. Typically, purities of 99.995% or higher are used. Trade names for typical purities include "Zero Grade," "Ultra-High Purity (UHP) Grade," "4.5 Grade" and "5.0 Grade."
The carrier gas flow rate affects the analysis in the same way that temperature does (see above). The higher the flow rate the faster the analysis, but the lower the separation between analytes. Selecting the flow rate is therefore the same compromise between the level of separation and length of analysis as selecting the column temperature.
With GCs made before the 1990s, carrier flow rate was controlled indirectly by controlling the carrier inlet pressure, or "column head pressure." The actual flow rate was measured at the outlet of the column or the detector with an electronic flow meter, or a bubble flow meter, and could be an involved, time consuming, and frustrating process. The pressure setting was not able to be varied during the run, and thus the flow was essentially constant during the analysis. The relation between flow rate and inlet pressure is calculated with Poiseuille's equation for compressible fluids.
Many modern GCs, however, electronically measure the flow rate, and electronically control the carrier gas pressure to set the flow rate. Consequently, carrier pressures and flow rates can be adjusted during the run, creating pressure/flow programs similar to temperature programs.
Inlet types and flow rates
The choice of inlet type and injection technique depends on if the sample is in liquid, gas, adsorbed, or solid form, and on whether a solvent matrix is present that has to be vaporized. Dissolved samples can be introduced directly onto the column via a COC injector, if the conditions are well known; if a solvent matrix has to be vaporized and partially removed, a S/SL injector is used (most common injection technique); gaseous samples (e.g., air cylinders) are usually injected using a gas switching valve system; adsorbed samples (e.g., on adsorbent tubes) are introduced using either an external (on-line or off-line) desorption apparatus such as a purge-and-trap system, or are desorbed in the S/SL injector (SPME applications).
Sample size and injection technique
The real chromatographic analysis starts with the introduction of the sample onto the column. The development of capillary gas chromatography resulted in many practical problems with the injection technique. The technique of on-column injection, often used with packed columns, is usually not possible with capillary columns. The injection system, in the capillary gas chromatograph, should fulfil the following two requirements:
#The amount injected should not overload the column.
#The width of the injected plug should be small compared to the spreading due to the chromatographic process. Failure to comply with this requirement will reduce the separation capability of the column. As a general rule, the volume injected, Vinj, and the volume of the detector cell, Vdet, should be about 1/10 of the volume occupied by the portion of sample containing the molecules of interest (analytes) when they exit the column.
Some general requirements, which a good injection technique should fulfill, are:
*It should be possible to obtain the column’s optimum separation efficiency.
*It should allow accurate and reproducible injections of small amounts of representative samples.
*It should induce no change in sample composition. It should not exhibit discrimination based on differences in boiling point, polarity, concentration or thermal/catalytic stability.
*It should be applicable for trace analysis as well as for undiluted samples.
Column temperature and temperature program
The column(s) in a GC are contained in an oven, the temperature of which is precisely controlled electronically. (When discussing the "temperature of the column," an analyst is technically referring to the temperature of the column oven. The distinction, however, is not important and will not subsequently be made in this article.)
The rate at which a sample passes through the column is directly proportional to the temperature of the column. The higher the column temperature, the faster the sample moves through the column. However, the faster a sample moves through the column, the less it interacts with the stationary phase, and the less the analytes are separated.
In general, the column temperature is selected to compromise between the length of the analysis and the level of separation.
A method which holds the column at the same temperature for the entire analysis is called "isothermal." Most methods, however, increase the column temperature during the analysis, the initial temperature, rate of temperature increase (the temperature "ramp") and final temperature is called the "temperature program."
A temperature program allows analytes that elute early in the analysis to separate adequately, while shortening the time it takes for late-eluting analytes to pass through the column.
Data reduction and analysis
Generally chromatographic data is presented as a graph of detector response (y-axis) against retention time (x-axis). This provides a spectrum of peaks for a sample representing the
analytes present in a sample eluting from the column at different times. Retention time can be used to identify analytes if the method conditions are constant. Also, the pattern of peaks will be constant for a sample under constant conditions and can identify complex mixtures of analytes. In most modern applications however the GC is connected to a mass spectrometer or similar detector that is capable of identifying the analytes represented by the peaks.
The area under a peak is proportional to the amount of analyte present. By calculating the area of the peak using the mathematical function of integration, the concentration of an analyte in the original sample can be determined. Concentration can be calculated using a
calibration curvecreated by finding the response for a series of concentrations of analyte, or by determining the relative response factorof an analyte. The relative response factor is the expected ratio of an analyte to an internal standard(or external standard) and is calculated by finding the response of a known amount of analyte and a constant amount of internal standard (a chemical added to the sample at a constant concentration, with a distinct retention time to the analyte).
In most modern
GC-MSsystems, computer software is used to draw and integrate peaks, and match MS spectra to library spectra.
In general, substances that vaporize below ca. 300 °C (and therefore are stable up to that temperature) can be measured quantitatively. The samples are also required to be
salt-free; they should not contain ions. Very minute amounts of a substance can be measured, but it is often required that the sample must be measured in comparison to a sample containing the pure, suspected substance.
Various temperature programs can be used to make the readings more meaningful; for example to differentiate between substances that behave similarly during the GC process.
Professionals working with GC analyze the content of a chemical product, for example in assuring the quality of products in the chemical industry; or measuring toxic substances in soil, air or water. GC is very accurate if used properly and can measure
picomoles of a substance in a 1 ml liquid sample, or parts-per-billion concentrations in gaseous samples.
In practical courses at colleges, students sometimes get acquainted to the GC by studying the contents of
Lavenderoil or measuring the ethylenethat is secreted by " Nicotiana benthamiana" plants after artificially injuring their leaves. These GC analyses hydrocarbons (C2-C40+). In a typical experiment, a packed column is used to separate the light gases, which are then detected with a TCD. The hydrocarbons are separated using a capillary column and detected with an FID. A complication with light gas analyses that include H2 is that He, which is the most common and most sensitive inert carrier (sensitivity is proportional to molecular mass) has an almost identical thermal conductivity to hydrogen (it is the difference in thermal conductivity between two separate filaments in a Wheatstone Bridge type arrangement that shows when a component has been eluted). For this reason, dual TCD instruments are used with a separate channel for hydrogen that uses nitrogen as a carrier are common. Argon is often used when analysing gas phase chemistry reactions such as F-T synthesis so that a single carrier gas can be used rather than 2 separate ones. The sensitivity is less but this is a tradeoff for simplicity in the gas supply.
GCs in popular culture
Movies, books and TV shows tend to misrepresent the capabilities of gas chromatography and the work done with these instruments.
In the U.S. TV show , for example, GCs are used to rapidly identify unknown samples. "This is
gasolinebought at a Chevron station in the past two weeks," the analyst will say fifteen minutes after receiving the sample.
In fact, a typical GC analysis takes much more time; sometimes a single sample must be run more than an hour according to the chosen program; and even more time is needed to "heat out" the column so it is free from the first sample and can be used for the next. Equally, several runs are needed to confirm the results of a study - a GC analysis of a single sample may simply yield a result per chance (see
Also, GC does not positively identify most samples; and not all substances in a sample will necessarily be detected. All a GC truly tells you is at which relative time a component eluted from the column and that the detector was sensitive to it. To make results meaningful, analysts need to know which components at which concentrations are to be expected; and even then a small amount of a substance can hide itself behind a substance having both a higher concentration and the same relative elution time. Last but not least it is often needed to check the results of the sample against a GC analysis of a reference sample containing only the suspected substance.
GC-MScan remove much of this ambiguity, since the mass spectrometerwill identify the component's molecular weight. But this still takes time and skill to do properly.
Similarly, most GC analyses are not
push-buttonoperations. You cannot simply drop a sample vial into an auto-sampler's tray, push a button and have a computer tell you everything you need to know about the sample. According to the substances one expects to find the operating program must be carefully chosen.
A push-button operation can exist for running similar samples repeatedly, such as in a chemical production environment or for comparing 20 samples from the same experiment to calculate the mean content of the same substance. However, for the kind of investigative work portrayed in books, movies and TV shows this is clearly not the case.
Thin layer chromatography
Gas chromatography-mass spectrometry
Unresolved Complex Mixture
* [http://www.gchelp.tk Gas Chromatography Help Site]
Wikimedia Foundation. 2010.
Look at other dictionaries:
gas-liquid chromatography — dujų skysčių chromatografija statusas T sritis Standartizacija ir metrologija apibrėžtis Dujų chromatografija, kai nejudanti fazė yra kietas nešiklis, padengtas darbinėje temperatūroje skystėjančiu junginiu. atitikmenys: angl. gas liquid… … Penkiakalbis aiškinamasis metrologijos terminų žodynas
gas-liquid chromatography — dujų skysčių chromatografija statusas T sritis chemija apibrėžtis Dujų chromatografija, kai nejudanti fazė yra kietas nešiklis, padengtas darbo temperatūroje skystėjančiu junginiu. atitikmenys: angl. gas liquid chromatography rus. газожидкостная… … Chemijos terminų aiškinamasis žodynas
gas-liquid chromatography — gas liq·uid chromatography gas lik wəd n gas chromatography in which the stationary phase is a liquid abbr. GLC gas liquid chromatographic adj * * * (GLC) see gas c … Medical dictionary
gas-liquid chromatography — noun Date: 1952 gas chromatography in which the stationary phase is a liquid • gas liquid chromatographic adjective … New Collegiate Dictionary
gas-liquid chromatography — ¦ ̷ ̷ ¦ ̷ ̷ ̷ ̷ noun : gas chromatography in which the stationary phase is a liquid • gas liquid chromatographic ¦ ̷ ̷ ¦ ̷ ̷ ̷ ̷ adjective … Useful english dictionary
gas-liquid chromatography — noun A form of chromatography in which the mobile phase is a gas and the stationary phase is a liquid, usually on small beads packed in a long column Syn: gas chromatography, gas, chromography … Wiktionary
gas–liquid chromatography — noun chromatography employing a gas as the moving carrier medium and a liquid as the stationary medium … English new terms dictionary
gas-liquid chromatography — /gæs ˌlɪkwəd kroʊməˈtɒgrəfi/ (say gas .likwuhd krohmuh togruhfee) noun → gas chromatography. Abbrev.: GLC … Australian English dictionary
gas-liquid chromatographic — adjective see gas liquid chromatography … New Collegiate Dictionary
gas-liquid chromatographic — adjective see gas liquid chromatography … Useful english dictionary