Fourier transform spectroscopy
Fourier transform spectroscopy is a measurement technique whereby spectra are collected based on measurements of the temporal coherence of a radiative source, using
time-domainmeasurements of the electromagnetic radiationor other type of radiation.It can be applied to a variety of types of spectroscopy including optical spectroscopy, infrared spectroscopy(IR), nuclear magnetic resonance, mass spectrometryand electron spin resonancespectroscopy. There are several methods for measuring the temporal coherence of the light, including the continuous wave "Michelson" or "Fourier transform" spectrometer and the pulsed Fourier transform spectrograph (which is more sensitive and has a much shorter sampling time than conventional spectroscopic techniques, but is only applicable in a laboratory environment).
Continuous wave "Michelson" or "Fourier transform" spectrograph
The Michelson spectrograph relies on the same principle as the
Michelson-Morley experiment. Light from the source is split into two beams by a half-silvered mirror, one is reflected off a fixed mirror and one off a moving mirror which introduces a time delay -- the Fourier transform spectrometer is just a Michelson interferometerwith a movable mirror. The beams interfere, allowing the temporal coherence of the light to be measured at each different time delay setting. By making measurements of the signal at many discrete positions of the moving mirror, the spectrum can be reconstructed using a Fourier transform of the temporal coherence of the light. Michelson spectrographs are capable of very high spectral resolution observations of very bright sources.The Michelson or Fourier transform spectrograph was popular for infra-red applications at a time when infra-red astronomy only had single pixel detectors. Imaging Michelson spectrometers are a possibility, but in general have been supplanted by imaging Fabry-Perotinstruments which are easier to construct.
Pulsed "Fourier transform" spectrometer
A pulsed "Fourier transform" spectrometer is usually used to measure the spectrum of the light transmitted through a laboratory sample. In a conventional (or "continuous wave") spectrometer, a sample is exposed to electromagnetic radiation and the response (usually the intensity of transmitted radiation) is monitored. The energy of the radiation is varied over the desired range and the response is plotted as a function of radiation energy (or frequency). At certain resonant frequencies characteristic of the specific sample, the radiation will be absorbed resulting in a series of peaks in the spectrum, which can then be used to identify the sample. (In magnetic spectroscopy, the
magnetic fieldis often varied insteadof the frequency of the incident radiation, though the spectra areeffectively the same as if the field had been kept constant and thefrequency varied. This is largely a question of experimentalconvenience.)
Instead of varying the wavelength of the electromagnetic radiation, Fourier Transform spectroscopy exposes the sample to a single pulse of radiation and measures the response.The resulting signal, called a "free induction decay," is a direct measurement of the temporal coherence of the light and contains a rapidly decaying composite of all possible frequencies. Using a Fourier transform of this, the spectrum of the light can be calculated as for the Michelson Fourier transform spectrometer.In this way the Fourier transform spectrometer can produce the same kind of spectrum as a conventional spectrometer, but in a much shorter time.The principles of the Fourier transform approach can be compared to the behavior of a musical
tuning fork.If a tuning fork is exposed to sound waves of varying frequencies, it will vibrate when the sound wave frequencies are in "tune".This is similar to conventional spectroscopic techniques, where the radiation frequency is varied and those frequencies where the sample is in "tune" with the radiation detected.However, if we strike the tuning fork (the equivalent of applying a pulse of radiation), the tuning fork will tend to vibrate at its characteristic frequencies.The resulting tone consists of a combination of all of the characteristic frequencies for that tuning fork.Similarly the response from a sample exposed to a pulse of radiation is a signal consisting primarily of the characteristic frequencies for that sample. The Fourier transform is a mathematical technique for determining these characteristic frequencies from a single composite signal.
attenuated total reflectance(ATR) crystalmakes it possible to increase the signal, facilitating surface analysis. It is normally done by pressing a flexible sample against the surface of a large seleniumcrystal.
One of the most important advantages of FTS was shown by P.B. Fellgett, an early advocate of the method. The Fellgett advantage, also known as the multiplex principle, states that a multiplex spectrometer such as the FTS will produce a gain of the order of the square root of m in the signal-to-noise ratio of the resulting spectrum, when compared with an equivalent scanning monochromator, where m is the number of elements comprising the resulting spectrum when the measurement noise is dominated by detector noise.
Converting spectra from time domain to frequency domain
:The sum is performed over all contributing frequencies to give a signal S(t) in the time domain.:gives non-zero value when S(t) contains a component that matches the oscillating function.
References and notes
Forensic polymer engineering
*cite journal | author = Ellis, D.I. and Goodacre, R. | title = Metabolic fingerprinting in disease diagnosis: biomedical applications of infrared and Raman spectroscopy | journal = The Analyst | year = 2006 | volume = 131 | pages = 875–885 | doi = 10.1039/b602376m
* [http://scienceworld.wolfram.com/physics/FourierTransformSpectrometer.html Description of how a Fourier transform spectrometer works]
* [http://www.astro.livjm.ac.uk/courses/phys362/notes/ The Michelson or Fourier transform spectrograph]
* [http://www.ijvs.com/volume5/edition5/section1.html#Feature Internet Journal of Vibrational Spectroscopy - How FTIR works]
* [http://www.osa.org/meetings/topicalmeetings/fts/default.aspx Fourier Transform Spectroscopy Topical Meeting and Tabletop Exhibit]
Wikimedia Foundation. 2010.
Look at other dictionaries:
Fourier-transform spectroscopy — Furjė spektroskopija statusas T sritis Standartizacija ir metrologija apibrėžtis Optinės spektroskopijos metodas, kuriuo spektras gaunamas dviem pakopomis: pirmiausiai registruojama tiriamosios spinduliuotės interferograma, po to Furjė keitimo… … Penkiakalbis aiškinamasis metrologijos terminų žodynas
Fourier-transform spectroscopy — Furjė spektroskopija statusas T sritis Standartizacija ir metrologija apibrėžtis Spektroskopijos metodas, kuriuo randamas spektras: pirmiausia užrašoma tiriamosios spinduliuotės interferograma, tada atliekama jos Furjė analizė. atitikmenys: angl … Penkiakalbis aiškinamasis metrologijos terminų žodynas
Fourier-transform spectroscopy — Furjė spektroskopija statusas T sritis fizika atitikmenys: angl. Fourier spectroscopy; Fourier transform spectroscopy vok. Fourier Spektroskopie, f rus. спектроскопия Фурье, f; Фурье спектроскопия, f pranc. spectroscopie de Fourier, f;… … Fizikos terminų žodynas
Fourier transform — Fourier transforms Continuous Fourier transform Fourier series Discrete Fourier transform Discrete time Fourier transform Related transforms The Fourier transform is a mathematical operation that decomposes a function into its constituent… … Wikipedia
Fourier transform ion cyclotron resonance — Infobox chemical analysis name = Fourier transform ion cyclotron resonance caption =A FTMS instrument at the Pacific Northwest National Laboratory, USA acronym = FTMS, FTICR classification =Mass spectrometry analytes = manufacturers = related =… … Wikipedia
Fourier transform — (FT) A mathematical operation which converts a function of one independent variable to one of a different independent variable. In FT–IR spectroscopy, the Fourier transform converts a time function (the interferogram) to a frequency function… … Forensic science glossary
Diffuse reflectance infrared fourier transform — Spectroscopy (DRIFTS) is an infrared spectroscopy spectra technique used on powder samples with no preparation. The sample is added to a sample cup and the data is collected on the bulk sample. The infrared light on a sample is reflected and… … Wikipedia
Fourier — (pronEng|ˈfʊərieɪ, French pronunciation IPA2|fuʁie) may refer to:*Charles Fourier (1772–1837), a French utopian socialist thinker *Joseph Fourier (1768–1830), a French mathematician and physicist **Mathematics, physics, and engineering terms… … Wikipedia
Spectroscopy — Analysis of white light by dispersing it with a prism is example of spectroscopy. Spectroscopy ( … Wikipedia
Fourier spectroscopy — Furjė spektroskopija statusas T sritis Standartizacija ir metrologija apibrėžtis Optinės spektroskopijos metodas, kuriuo spektras gaunamas dviem pakopomis: pirmiausiai registruojama tiriamosios spinduliuotės interferograma, po to Furjė keitimo… … Penkiakalbis aiškinamasis metrologijos terminų žodynas