# Disc permeameter

The disc permeameter is a field instrument used for measuring water infiltration in the soil, which is characterized by in situ saturated and unsaturated soil hydraulic properties. It is mainly used to provide estimates of the hydraulic conductivity of the soil near saturation.

## History

Conventional techniques for measuring in-situ infiltration include the use of a single or double ring infiltrometer. Single and double ring infiltrometer only measures flow under ponded (saturated) conditions, and when used in soil with distinct macropores, preferential flow will dominate the flow. (See: Poiseuille's law) This does not reflect infiltration under rainfall or sprinkler irrigation. Therefore many authors attempted to create a negative potential (tension) on the water flow. This is to exclude macropores in the flow process, hence only measuring the soil matrix flow.

The CSIRO disc permeameter for tension (left) and ponded (right) infiltration

Willard Gardner and Walter Gardner developed a negative head permeameter as early as 1939. Dixon (1975) developed a closed-top ring infiltrometer to quantify macropores. Water is applied to a closed-top system, which permits the imposition of negative head or pressure on the ponded water surface. Negative tension can be considered as simulating a positive soil air pressure, created by a negative air pressure above ponded surface water. A simplification was made by Topp and Zebchuk (1985). The limitation of this device is the infiltration has to be started by ponding the closed-top infiltrometer (applying a positive head), then adjusted to a negative pressure. Little research effort was continued in this area, instead attention has been given mainly to the sorptivity apparatus of Dirksen (1975) which used a ceramic plate as a base. Based on this design, Brent Clothier and Ian White (1981) developed the sorptivity tube which can provide a constant negative potential (tension) on the soil surface. However, the sorptivity tube had many shortcomings, hence modifications to the design led to the development of the disc permeameter by Perroux and White (1988) from CSIRO. In the US it is known as the tension infiltrometer.

For more on the development of the first permeameter as told by Walter Gardner, visit (http://www.decagon.com/ag_research/hydro/history.php)

## The Disc

The CSIRO disc permeameter of Perroux and White (1988) (not patented) comprises a nylon mesh supply membrane (with a very small diameter around 10-40 mm), a water reservoir and a bubbling tower. The bubbling tower is connected to the reservoir and is open to air. The bubbling tower controls the potential h0 applied to the membrane by adjusting the water height in the air-inlet tube. So essentially the soil pores need to have energy equivalent to h0 to overcome water that is held under tension in the reservoir. It can be used to supply potential ranging -200 mm to 0 mm, effectively excluding pores with diameter bigger than 0.075 mm.

Many different designs have evolved, including:

## Mathematical analysis

The CSIRO ponded disc permeameter

Due to the three-dimensional water flow from the disc, a special formulation is needed to take into account the lateral absorption of water. The analyses are derived from the simple, steady-state analysis of Wooding (1968). For steady infiltration from a circular, shallow, inundated area, Wooding found that a remarkable feature of this curve is the fact that it never departs far from the straight line:

Q * = 2πa + 4

where Q* is the dimensionless flux, a = αr / 2. r is the radius of the disc [cm] and α [1/cm] is the sorptive number or the parameter of Gardner’s (1958) hydraulic conductivity function:

K(h) = Kseαh

where K is the hydraulic conductivity [cm/h], Ks is saturated conductivity and h is soil water potential [cm]. In terms of the actual steady-state infiltration rate q¥ [cm/h]:

$q_\infty = \alpha \phi_0 + \frac{4 \phi_0}{\pi_r}$

## References

• Clothier, B.E., White, I., 1981. Measurement of sorptivity and soil water diffusivity in the field. Soil Science Society of America Journal 45, 241-245.
• Dirksen, C., 1975. Determination of soil water diffusivity by sorptivity measurements. Soil Science Society of America Proceedings 39, 22-27.
• Dixon, R.M., 1975. Design and use of closed-top infiltrometers. Soil Science Society of America Proceedings 39, 755-763.
• Perroux, K.M., White, I., 1988. Design for disc permeameters. Soil Science Society of America Journal, 52, 1205-1215.
• Wooding, R.A., 1968. Steady infiltration from a shallow circular pond. Water Resources Research 4, 1259-1273.

Wikimedia Foundation. 2010.

### Look at other dictionaries:

• Characterisation of pore space in soil — Soil is essential to most animals on the earth. It is a relatively thin crust where an even smaller portion contains much of the biological activity. Soil consists of three different phases. A solid phase (≈ 20%) that contains mainly minerals of… …   Wikipedia

• Infiltrometer — is a device used to measure the rate of water infiltration into soil or other porous media. Commonly used infiltrometers are single ring or double ring infiltrometer. Others are the disc permeameter, and the sprinkler infiltrometer.The single… …   Wikipedia

• Macropore — Macropores are large diameter (more than 50 nm, see macroporous materials) conduits in the soil, created by such agents as plant roots, soil cracks, or soil fauna. Macropores increase the hydraulic conductivity of the soil, allowing water to… …   Wikipedia

• Outline of hydrology — The following outline is provided as an overview of and topical guide to hydrology: Hydrology – study of the movement, distribution, and quality of water on Earth and other planets, including the hydrologic cycle, water resources and… …   Wikipedia