Square Section Autoclave
A modern front-loading autoclave
Uses Sterilization Inventor Charles Chamberland Manufacturer Astell Scientific Related items Waste autoclave
An autoclave is an instrument to sterilize equipment and supplies by subjecting them to high pressure saturated steam at 121 °C for around 15–20 minutes depending on the size of the load and the contents. It was invented by Charles Chamberland in 1879, although a precursor known as the steam digester was created by Denis Papin in 1679. The name comes from Greek auto-, ultimately meaning self, and Latin clavis meaning key — a self-locking device.
Autoclaves are widely used in microbiology, medicine, tattooing, body piercing, veterinary science, mycology, dentistry, chiropody and prosthetic fabrication. Many autoclaves vary in size and function depending on the media they are sterilizing.
A notable growing application of autoclaves is in the pre-disposal treatment and sterilization of waste material, such as pathogenic hospital waste. Machines in this category largely operate under the same principles as the original autoclave in that they are able to neutralize potentially infectious agents by utilizing pressurized steam and superheated water. A new generation of waste converters is capable of achieving the same effect without any pressure vessels to sterilize culture media, rubber material, gowns, dressing, gloves etc. It is particularly useful for materials which cannot withstand the higher temperature of a hot air oven. For all-glass syringes, a hot air oven is a better sterilizing method.
Autoclaves are also widely used to cure composites and in the vulcanization of rubber. The high heat and pressure that autoclaves allow help to ensure that the highest possible physical properties are repeatably attainable. Aerospace and sparmakers (for sailboats in particular) have autoclaves well over 50 feet long, some over 10 feet wide.
It is very important to ensure that all of the trapped air is removed, as hot air is very poor at achieving sterility. Steam at 134 °C can achieve in 3 minutes the same sterility that hot air at 160 °C takes two hours to achieve. Methods of achieving air removal include:
Downward displacement (or gravity type) - As steam enters the chamber, it fills the upper areas as it is less dense than air. This compresses the air to the bottom, forcing it out through a drain. Often a temperature sensing device is placed in the drain. Only when air evacuation is complete should the discharge stop. Flow is usually controlled through the use of a steam trap or a solenoid valve, but bleed holes are sometimes used, often in conjunction with a solenoid valve. As the steam and air mix it is also possible to force out the mixture from locations in the chamber other than the bottom.
Steam pulsing - Air dilution by using a series of steam pulses, in which the chamber is alternately pressurized and then depressurized to near atmospheric pressure.
Vacuum pumps - Vacuum pumps to suck air or air/steam mixtures from the chamber.
Superatmospheric uses a vacuum pump. It starts with a vacuum followed by a steam pulse and then a vacuum followed by a steam pulse. The number of pulses depends on the particular autoclave and cycle chosen.
Subatmospheric - Similar to superatmospheric cycles, but chamber pressure never exceeds atmospheric until they pressurize up to the sterilizing temperature.
Autoclaves in medicine
A medical autoclave is a device that uses steam to sterilize equipment and other objects. This means that all bacteria, viruses, fungi, and spores are inactivated. However, prions, like those associated with Creutzfeldt-Jakob disease, may not be destroyed by autoclaving at the typical 134 °C for 3 minutes or 121 °C for 15 minutes. Also, some recently-discovered organisms, such as Strain 121, can survive at temperatures above 121 °C.
Autoclaves are found in many medical settings, laboratories, and other places that need to ensure sterility of an object. Many procedures today use single-use items rather than sterilized, reusable items. This first happened with hypodermic needles, but today many surgical instruments (such as forceps, needle holders, and scalpel handles) are commonly single-use items rather than reusable (see waste autoclave).
Because damp heat is used, heat-labile products (such as some plastics) cannot be sterilized this way or they will melt. Some paper or other products that may be damaged by the steam must also be sterilized another way. In all autoclaves, items should always be separated to allow the steam to penetrate the load evenly.
Autoclaving is often used to sterilize medical waste prior to disposal in the standard municipal solid waste stream. This application has grown as an alternative to incineration due to environmental and health concerns raised by combustion byproducts from incinerators, especially from the small units which were commonly operated at individual hospitals. Incineration or a similar thermal oxidation process is still generally mandated for pathological waste and other very toxic and/or infectious medical wastes.
Autoclave quality assurance
Chemical indicators can be found on medical packaging and autoclave tape, and these change color once the correct conditions have been met. This color change indicates that the object inside the package, or under the tape, has been processed. Biological indicators contain spores of a heat-resistant bacterium, Geobacillus stearothermophilus. If the autoclave does not reach the right temperature, when incubated the spores will germinate, and their metabolism will change the color of a pH-sensitive chemical. Some physical indicators consist of an alloy designed to melt only after being subjected to the relevant holding time. If the alloy melts, the change will be visible.
Some computer-controlled autoclaves use an F0 (F-nought) value to control the sterilization cycle. F0 values are set as the number of minutes of equivalent sterilization at 121 °C (250 °F) at 15 psi (100 kPa) above atmospheric pressure for 15 minutes . Since exact temperature control is difficult, the temperature is monitored, and the sterilization time adjusted accordingly.
- ^ Microbiology, Jacquelyn Black, Prentice Hall,1993 pg 334
- ^ "Chronological reference marks - Charles Chamberland (1851–1908)". Pasteur Institute. http://www.pasteur.fr/infosci/archives/chb0.html. Retrieved 2007-01-19.
- ^ Hugo WB (July 1991). "A brief history of heat and chemical preservation and disinfection". J. Appl. Bacteriol. 71 (1): 9–18. doi:10.1111/j.1365-2672.1991.tb04657.x. PMID 1894581. http://www3.interscience.wiley.com/cgi-bin/fulltext/119995683/PDFSTART.
- ^ http://www.etymonline.com/index.php?term=autoclave
- ^ "Sterilization Cycles". Consolidated Machine Corporation. http://consteril.com/index.php?pg=41. Retrieved 2009-06-30.
- ^ AS NZS 4815-2006 P33&P35
Laboratory equipment Glassware
Beaker · Boston round (bottle) · Büchner funnel · Burette · Cold finger · Condenser · Conical measure · Cuvette · Dean-Stark apparatus · Dropping funnel · Eudiometer · Evaporating dish · Gas syringe · Graduated cylinder · Pipette · Petri dish · Pycnometer · Separatory funnel · Soxhlet extractor · Watch glass
Agar plate · Aspirator · Autoclave · Biosafety cabinet · Bunsen burner · Calorimeter · Chemostat · Colony counter · Colorimeter · Laboratory centrifuge · Crucible · Eyewash · Fire blanket · Fume hood · Glove box · Homogenizer · Hot air oven · Incubator · Laminar flow cabinet · Magnetic stirrer · Meker-Fisher burner · Microscope · Microtiter plate · Picotiter plate · Plate reader · Retort stand · Safety shower · Spectrophotometer · Static mixer · Stir bar · Stirring rod · Scoopula · Teclu burner · Thermometer · Vacuum dry box · Vortex mixer · Wash bottleSee also Instruments used in medical laboratories
Wikimedia Foundation. 2010.
Look at other dictionaries:
Autoclave — moderne de laboratoire … Wikipédia en Français
autoclave — [ otoklav ] adj. et n. m. • 1820; de auto et lat. clavis « clé » ♦ Qui se ferme de soi même. Appareil, marmite autoclave. ♢ N. m. Récipient métallique à fermeture extérieure hermétique, résistant à des pressions élevées. ⇒ étuve. Stérilisation de … Encyclopédie Universelle
Autoclave — Saltar a navegación, búsqueda Autoclave puede referirse a: Autoclave de uso médico usado para esterilizar instrumental y otro producto sanitario. Autoclave de laboratorio usado para esterilizar material de laboratorio. Autoclave industrial como… … Wikipedia Español
Autoclave — Основная информация … Википедия
autoclave — f. microb. Aparato utilizado para esterilizar materiales mediante calor húmedo a presión. Consta de un recipiente de gruesas paredes metálicas que permite alcanzar altas presiones. Medical Dictionary. 2011. autoclave … Diccionario médico
autoclave — ‘Aparato para la esterilización por vapor a presión y altas temperaturas’. Se usa en ambos géneros, con predominio hoy del masculino: «La esterilización se realiza en el autoclave» (Pozuelo/PzPérez Técnicas [Esp. 2001]); «Las autoclaves… … Diccionario panhispánico de dudas
Autoclave — Au to*clave, n. [F., fr. Gr. a yto s self + L. clavis key.] 1. A kind of French stewpan with a steam tight lid. Knight. [1913 Webster] 2. a device used for sterilizing objects by exposing them to steam at above atmospheric pressure (and thus at a … The Collaborative International Dictionary of English
autoclave — 1880, from French, lit. self locking, from auto self (see AUTO (Cf. auto )) + L. clavis key (see SLOT (Cf. slot) (2)) … Etymology dictionary
autoclave — s. f. Aparelho de pressão de vapor empregado na esterilização de instrumentos cirúrgicos, pensos, etc … Dicionário da Língua Portuguesa
autoclave — sustantivo femenino 1. Área: tecnología Aparato de cierre hermético para la esterilización de objetos por medio de vapor a alta presión … Diccionario Salamanca de la Lengua Española
autoclave — (De auto y clave). f. Aparato que sirve para esterilizar objetos y sustancias situados en su interior, por medio de vapor y altas temperaturas … Diccionario de la lengua española