A thermographic camera, sometimes called a
FLIR(Forward Looking InfraRed), or an infrared camera less specifically, is a device that forms an image using infrared radiation, similar to a common camerathat forms an image using visible light. Instead of the 450–750 nanometer range of the visible light camera, infrared cameras operate in wavelengths as long as 14,000 nm (14 µm).
Theory of operation
Infrared energy is just one part of the
electromagnetic spectrumthat encompasses radiation from gamma rays, x-rays, ultra violet, a thin region of visible light, infrared, microwaves, and radio waves. These are all related and differentiated in the length of their wave (wavelength). All objects emit a certain amount of black bodyradiation as a function of their temperatures. Generally speaking, the higher an object's temperature is, the more infrared radiation as black-body radiation it emits. A special cameracan detect this radiation in a way similar to an ordinary camera does visible light. It works even in total darkness because ambient light level does not matter. This makes it useful for rescue operations in smoke-filled buildings and underground.
Images from infrared cameras tend to be monochromatic because the cameras are generally designed with only a single type of sensor responding to single
wavelengthrange of infrared radiation. Color cameras require a more complex construction to differentiate wavelength and color has less meaning outside of the normal visible spectrum because the differing wavelengths do not map uniformly into the system of color vision used by humans. Sometimes these monochromatic images are displayed in pseudo-color, where changes in color are used rather than changes in intensity to display changes in the signal. This is useful because although humans have much greater dynamic range in intensity detection than color overall, the ability to see fine intensity differences in bright areas is fairly limited. This technique is called density slicing.
For use in temperature measurement the brightest (warmest) parts of the image are customarily colored white, intermediate temperatures reds and yellows, and the dimmest (coolest) parts blue. A scale should be shown next to a false color image to relate colors to temperatures. Their resolution is considerably lower than of optical cameras, mostly only 160x120 or 320x240 pixels. Thermographic cameras are much more expensive than their visible-spectrum counterparts, and higher-end models are often deemed as
Thermal imaging photography finds many other uses. For example,
firefighters use it to see through smoke, find persons, and localize hotspots of fires. With thermal imaging, power line maintenance technicians locate overheating joints and parts, a telltale sign of their failure, to eliminate potential hazards. Where thermal insulationbecomes faulty, building constructiontechnicians can see heat leaks to improve the efficiencies of cooling or heating air-conditioning. Thermal imaging cameras are also installed in some luxury cars to aid the driver, the first being the 2000 Cadillac DeVille. Some physiological activities, particularly responses, in human beings and other warm-blooded animals can also be monitored with thermographic imaging. Cooled infrared cameras can also be found at most major astronomy research telescopes.
Thermographic cameras can be broadly divided into two types: those with cooled infrared image detectors and those with uncooled detectors.
Cooled infrared detectors
Cooled detectors are typically contained in a vacuum-sealed case or flask and
cryogenically cooled. This greatly increases their sensitivitysince their own temperatures are much lower than that of the objects from which they are meant to detect radiation. Typical cooling temperatures range from 4 K to 110 K, 80 K to 90 K being the most common. Without cooling, these sensors (which detect and convert light in much the same way as common digital cameras, but are made of different materials) would be 'blinded' or flooded by their own radiation. The drawbacks of cooled infrared cameras are that they are expensive both to produce and to run. Cooling and evacuating are power- and time-consuming. The camera may need several minutes to cool down before it can begin working. Although the components that lower temperature and pressure are generally bulky and expensive, cooled infrared cameras provide superior image quality compared to uncooled ones. One approach is to use high pressure nitrogen cleaned to a very high standard (white spot 6-> 7 equivalent). The pressurised gas is expanded via a micro-sized orifice and passed over a miniature heat exchanger resulting in regenerative cooling via the Joule Thomson effect of the expanding gas cooling & heat exchanger combination. It is not difficult to imagine the cost of creating, storing and deploying such a system.
Materials used for infrared detection include
liquid heliumcooled silicon bolometers, and a wide range of cheaper narrow gap semiconductordevices including:
In principle, superconducting tunneling junction devices could be used as well as infrared sensors because of their very narrow gap. Small arrays have been demonstrated. Their wide range use is difficult because their high sensitivity requires careful shielding from the background radiation.
Uncooled infrared detectors
Uncooled thermal cameras use a sensor operating at ambient temperature, or a sensor stabilized at a temperature close to ambient using small temperature control elements. Modern uncooled detectors all use sensors that work by the change of resistance,
voltageor current when heated by infrared radiation. These changes are then measured and compared to the values at the operating temperature of the sensor. Uncooled infrared sensors can be stabilized to an operating temperature to reduce image noise, but they are not cooled to low temperatures and do not require bulky, expensive cryogenic coolers. This makes infrared cameras smaller and less costly. However, their resolution and image quality tend to be lower than cooled detectors. This is due to difference in their fabricational processes, limited by currently available technology.
Uncooled detectors are mostly based on
pyroelectricand ferroelectricmaterials [http://oemagazine.com/fromthemagazine/jun01/hotdetectors.html] or microbolometertechnology.
Some of the materials used for the sensor arrays are eg.: [http://www.freepatentsonline.com/5990481.html]
vanadium(V) oxide(metal insulator phase change material, for microbolometer arrays)
lanthanum barium manganite(LBMO, metal insulator phase change material)
lead zirconate titanate(PZT)
lanthanum dopedlead zirconate titanate (PLZT)
lead scandium tantalate(PST)
lead lanthanum titanate(PLT)
lead zinc niobate(PZN)
lead strontium titanate(PSrT)
barium strontium titanate(BST)
Aside from test equipment, training is the most important investment a company will make in an infrared inspection program. Advances in technology have provided infrared equipment that is user-friendly; however, infrared thermography is not a "simply point and shoot" technology. In addition to understanding the object or system being inspected, thermographers must also understand common error sources that can influence observed thermal data. Typically, infrared training courses should cover the topics of infrared theory, heat transfer concepts, equipment selection and operation, how to eliminate or overcome common error sources, and specific applications.
Originally developed for military use during the
Korean War, thermographic cameras have slowly migrated into other fields as varied as medicine and archeology. More recently, the lowering of prices have helped fuel the adoption of infrared viewing technology. Advanced optics and sophisticated software interfaces continue to enhance the versatility of IR cameras.
* Astronomy, in devices such as the
Spitzer Space Telescope
* Firefighting operations
* Military & Police target detection & acquisition
* Law enforcement and anti-terrorism
Predictive maintenance(early failure warning) on mechanical & electrical equipment
* Process monitoring
* Condition Monitoring & surveillance
* Automotive applications
* Energy auditing of
building insulationand detection of refrigerantleaks
* Roof inspection
* Auditing of acoustic insulation for sound reduction
* Masonry wall structural analysis
* Moisture detection in walls & roofs
* Medical testing for diagnosis
Quality controlin production environments
* Research & development of new products
* Locating unmarked graves
Search and rescueoperations
Technical Surveillance Counter-Measures
Quarantinemonitoring of visitors to a country
Some specification parameters of an infrared camera system are:
* Number of
* Spectral band
* Sensor life time
Minimum resolvable temperature difference(MRTD)
Field of view
* Input power
* Mass and volume
Digital infrared thermal imaging in health care
Thermal imaging camera
* [http://www.irinfo.org/Articles/articles.html IRINFO.ORG - The Definitive Online Resource for Infrared Thermography]
* [http://www.buildscanir.com/modules.php?name=Content&pa=showpage&pid=1 BuildScanIR Network - Providing expert advice, training, support, and services to infrared thermographers]
* [http://vzone.virgin.net/ljmayes.mal/pubs/uncooled.htm Uncooled thermal imaging works round the clock]
* [http://www.grayess.com/software Standard thermographic analysis software supporting many camera models]
Wikimedia Foundation. 2010.
Look at other dictionaries:
Thermographic inspection — refers to the nondestructive testing of parts, materials or systems through the imaging of the thermal patterns at the object s surface. Strictly speaking, the term thermography alone, refers to all thermographic inspection techniques regardless… … Wikipedia
Thermal imaging camera — A thermal imaging camera (colloquially known as a TIC) is a specific type of thermographic camera used in firefighting. By rendering infrared radiation as visible light, they allow firefighters to see areas of heat through smoke, darkness, or… … Wikipedia
Thermography — Infrared Thermography, thermal imaging, or thermal video, is a type of infrared imaging science. Thermographic cameras detect radiation in the infrared range of the electromagnetic spectrum (roughly 900 ndash;14,000 nanometers or 0.9 ndash;14 µm) … Wikipedia
Infrared — For other uses, see Infrared (disambiguation). An image of two people in mid infrared ( thermal ) light (false color) … Wikipedia
Automotive night vision — An automotive night vision system is a system to increase a vehicle driver s perception and seeing distance in darkness or poor weather beyond the reach of the vehicle s headlights. They are currently offered as optional equipment on certain… … Wikipedia
Night vision — For other uses, see Night vision (disambiguation). Two American soldiers pictured during the 2003 Iraq War seen through an image intensifier Night vision is the ability to see in low light conditions. Whether by biological or technological means … Wikipedia
Machine vision glossary — Common definitions related to the machine vision field.Compiled for application on the Society of Manufacturing Engineers interest area.General related fields *Machine vision *Computer vision *Image processing *Signal processing NOTOC 0 9*1394.… … Wikipedia
Cutting extinguisher — The use of a cutting extinguisher is a fire extinguishing technique that combines abrasive waterjet cutting with water spray extinguishing, through a single handpiece or nozzle. The fire fighter approaches the fire from outside the main fire area … Wikipedia
MythBusters (2008 season) — Country of origin Australia United States No. of episodes 20 (includes 8 specials) Broadcast Original channel Discovery Channel … Wikipedia
German Aerospace Center — Deutsches Zentrum für Luft und Raumfahrt e.V. Established 1969 Headquarters Köln Porz Administrator Johann Dietrich Wörner … Wikipedia