- Continuity tester
A continuity tester is an item of electrical test equipment used to determine if an electrical path can be established between two points; that is if an electrical circuit can be made. The circuit under test is completely de-energized prior to connecting the apparatus.
The tester consists of an indicator in series with a source of electrical power - normally a battery, terminating in two test leads. If a complete circuit is established between the test-leads, the indicator is activated.
The indicator may be an electric light or such a device might also be fitted with a buzzer. This led to the term "buzzing out a circuit" (which means to test for continuity). This term remains in use today, although any sound produced is usually made by the beeper function on a modern digital voltmeter, but the most simple method that has been used for many years, is a simple curcuit consisting of a 9V battery and a buzzer, with a break in the curcuit to touch what is to be tested at each end to create a complete curcuit and to set off the buzzer if no fault is present.
A popular design has the tester combined with a standard flashlight. A TRS connector or jack plug in the rear of the unit permits a set of test leads to be plugged in effecting a quick conversion between the two applications.
For situations where continuity testing must be done on high resistance circuits, or where delicate conductors and sensitive components that might be damaged by excessive current are present, a low voltage, low current device must be used. These typically use an op-amp and watch batteries to drive an LED as an indicator. These testers can be exquisitely sensitive; for example they will indicate if the test points are taken by both hands.
There are times when a simple continuity test fails to reveal the problem. For example, vibration-induced problems in automobile wiring can be extremely difficult to detect because a short or open is not maintained long enough for a standard tester to respond.
In these applications a latching continuity tester is used. A more complex device, it detects intermittent opens and shorts as well as steady-state conditions. These devices contain a fast acting electronic switch (generally a Schmitt trigger) forming a gated astable oscillator which detects and locks (latches) the indicator on an intermittent condition with a duration of less than a millisecond.
To buzz out, a curcuit is usually the first thing that anyone in electrics, when testing cables, will do to locate the break or fault in the connection, and in doing so, see if it is actually the cable or the connecting component that has created the fault.
- ^ a b c d "Cable Continuity Tester Info Source". http://www.continuitytester.net/cable-continuity-tester-info-source.html. Retrieved April 28, 2011.
- ^ a b c Dean, Tamara (2010). Network+ Guide to Networks. Delmar. pp. 660–661. http://www.coursesmart.com/network-guide-to-networks-5th-edition/dean/dp/9781423902454.
- ^ a b Dean, Tamara (2010). Network+ Guide to Networks. Delmar. pp. 661–662.
Wikimedia Foundation. 2010.
Look at other dictionaries:
Cable tester — A cable tester is an electronic device used to verify the electrical connections in a cable or other wired assembly. Generally a cable tester consists of: # A source of electric current, # A volt meter, # A switching matrix used to connect the… … Wikipedia
circuit tester — A tool which looks like a screwdriver with a light at the end of the handle as well as a long wire with an allegator clip. The pointed end touches the hot wire while the allegator end touches or clips to the ground. If there is continuity and… … Dictionary of automotive terms
Test light — Neon test lamp for line voltages A test light, test lamp, voltage tester, or mains tester is a very simple piece of electronic test equipment used to determine the presence or absence of an electric voltage in a piece of equipment under test.… … Wikipedia
Solenoid voltmeter — A solenoid voltmeter is a specific type of voltmeter used by electricians in the testing of electrical power circuits. Wiggy is the registered trademark for a common solenoid voltmeter used in North America and manufactured by Square D, derived… … Wikipedia
Copper cable certification — 10G Certification Kit In copper twisted pair wire networks, copper cable certification is achieved through a thorough series of tests in accordance with Telecommunications Industry Association (TIA) or International Organization for… … Wikipedia
Electronic test equipment — (sometimes called testgear ) is used to create signals and capture responses from electronic Devices Under Test (DUTs). In this way, the proper operation of the DUT can be proven or faults in the device can be traced and repaired. Use of… … Wikipedia
electrical components — ▬▬▬▬▬▬▬▬ Types of electrical components and devices include: adaptor ammeter armature battery bayonet fitting cable ceiling rose circuit breaker conduit continuity tester copper conductor dimmer switch dry cell battery earthed plug electrical… … Useful english dictionary
LED circuit — In electronics, the basic LED circuit is an electrical circuit used to power a light emitting diode (LED). It consists of up to four components connected in series: a voltage source, a current limiting resistor, an LED, and optionally a switch to … Wikipedia
прибор для контроля целостности изоляции проводов — — [А.С.Гольдберг. Англо русский энергетический словарь. 2006 г.] Тематики энергетика в целом EN continuity tester … Справочник технического переводчика
тестер для проверки цепей на разрыв — — [Я.Н.Лугинский, М.С.Фези Жилинская, Ю.С.Кабиров. Англо русский словарь по электротехнике и электроэнергетике, Москва, 1999 г.] Тематики электротехника, основные понятия EN continuity tester … Справочник технического переводчика