Network data continues to be transmitted across an electrically non-conducting barrier, through the applied principle of electromagnetic induction, whereby high frequency AC voltages conveying data are induced across an isolating gap.
The network isolator is therefore a passive device, and functions without any requirement of an external power supply.
Network isolators are used in the medical industry to protect patients against leakage currents.
Network connections between medical devices and Ethernet networks (and other non-medical devices, such as personal computers) must be in accordance with the IEC standard IEC 60601-1 (3rd edition). This standard specifically deals with medical electronic and electrical equipment and systems, and classifies non-medical devices as potential hazard sources. A specific hazard stems from possible differences in ground potentials between network components, which when not properly isolated can result in a leakage current that can flow through the patient, which is dangerous, and potentially lethal. Such voltage differences can also arise through incorrect installation and wiring of network systems, electrical shorting within damaged cables and cabling, or shorting between damaged network cables and other voltage sources.
Network isolators work to remove this hazard, by electrically disconnecting medical devices from a network. Isolators may be used as network accessories, built into medical devices, or installed within a medical network system. Networks isolators should be installed as close as possible to the medical device in question.
As they serve no therapeutic or diagnostic purpose, network isolators themselves are not classed as medical electronic equipment according to the standard IEC 60601-1, nor do they fall within the scope of the Medical Devices Directive 93/42/EEC.
They are often installed in the medical field in conjunction with isolation transformers, which serve to protect the patient from electrical faults.
Network isolators have many other applications in overcoming the problems of differing ground potentials across networks, or between network components. For example:
- Electrical testing facilities, where measurement and monitoring equipment are connected to a central control centre through an Ethernet;
- For redundant server systems, which are physically connected via a copper network;
- For private or commercial networks, where inherent potential differences within a building, or between buildings, become problematic, and a fibre-optic solution is not economically viable;
- General usage as filters for shield currents in Ethernet connections.
The international standard IEC 60601-1 Medical Electrical Equipment (3rd edition) specifies stringent criteria on the safety and isolation of medical devices. These requirements are of course much more stringent than those of a typical consumer product standard (i.e. IEC 60950 Safety of Information Technology Equipment), as the patient may be unconscious, anaesthetised, or otherwise unable to move, and the need for patient protection is therefore paramount.
The IEC 60601-1 standard requires that two independent precautionary measures (Means of Patient Protection, or MOPPs) are set in place, to protect the patient from electrical shock in the medical system. A network isolator can be constructed to provide one or both required MOPPs. In the case where only a single MOPP is present in a system, a second measure of protection for the medical device must be made. This can be achieved, for example, by installing a fixed, low-resistance earthing cable, bar, or strap, referred to as a potential equalisation conductor. Conformation to the requirements of applicable standards and guidelines, in particular the IEC 60601-1 standard, involves manufacturers themselves declaring CE compliance and/or gaining certification from an approved independent testing laboratory.
The number of MOPPs a network isolator provides will be stated in the certification of the device. These are achieved by a combination of clearance and creepage distances, and the breakdown voltage of the insulating materials. Although requirements on breakdown voltage levels are 1.5 kilovolts for one MOPP, and 4 kilovolts for two MOPPs, the number of MOPPs actually provided cannot be derived from the breakdown voltage alone.
Network connections made with unshielded cables are occasionally used as a form of isolation, in that no shield connection exists between the device and the network. However, this fails to provide sufficient protection in the scope of IEC 60601-1, as the data lines themselves are not isolated, and may be floating at dangerous voltage potentials.
Network cards and other active network components which are designed to meet IEC 60950 (Safety of Information Technology Equipment), will usually not comply with the more demanding isolation requirements of IEC 60601-1 (Medical Electrical Equipment), and therefore require additional isolation before being used in medical applications.
By definition, a network isolator completely eliminates each and every direct electrical connection between devices connected through a network connection - specifically all the data lines, and the cable shield.
Network isolators are commercially available in various designs, to suit customers’ requirements:
- stand-alone devices, ready for immediate and simple installation;
- isolating components, suitable for OEM manufacturers;
- integrated into network wall sockets, for permanent installations.
Voltage ratings (rated in kilovolts) and maximum transmission speeds (e.g. 1000BASE-T) vary between models.
- Krankenhaus IT Journal Ausgabe 05/2008
- DeviceMed Ausgabe 10/2008
Wikimedia Foundation. 2010.
Look at other dictionaries:
Battery isolator — A battery isolator is a one way electrical valve, allowing DC current to flow in one direction, but not flow in reverse. They are commonly used on vehicles where multiple batteries or battery banks are used, including recreational vehicles, boats … Wikipedia
Ethernet over twisted pair — cable (upper) and 8P8C plug (lower) Ethernet over twisted pair technologies use twisted pair cables for the physical layer of an Ethernet computer network. Other Ethernet cable standards employ coaxial cable or optical fiber. Early versions… … Wikipedia
Reciprocity (electromagnetism) — This page is about reciprocity theorems in classical electromagnetism. See also Reciprocity (mathematics) for unrelated reciprocity theorems, and Reciprocity for more general usages of the term. In classical electromagnetism, reciprocity refers… … Wikipedia
List of astronomy acronyms — This is a compilation of acronyms commonly used in astronomy. Most of the acronyms are drawn from professional astronomy and are used quite frequently in scientific publications. However, a few of these acronyms are frequently used by the general … Wikipedia
SNI — System News, Inc. (Business » Firms) * Seneca Nation of Indians (Governmental » State & Local) * Sports Nutrition International (Community » Non Profit Organizations) * SNA Network Interconnection (Computing » Networking) * San Nicolas Island… … Abbreviations dictionary
Switch — For other uses, see Switch (disambiguation). Electrical switches. Top, left to right: circuit breaker, mercury switch, wafer switch, DIP switch, surface mount switch, reed switch. Bottom, left to right: wall switch (U.S. style), miniature toggle… … Wikipedia
Base station — The term base station can be used in the context of land surveying, wireless computer networking, and wireless communications.Land surveyingIn the context of external land surveying, a base station is a GPS receiver at an accurately known fixed… … Wikipedia
Electronic component — Various components An electronic component is a basic electronic element and may be available in a discrete form having two or more electrical terminals (or leads). These are intended to be connected together, usually by soldering to a printed… … Wikipedia
Recreational vehicle — This article is about the vehicle. For the film, see RV (film). RV redirects here. For other uses, see RV (disambiguation). A Class A motorhome with slide out extended floors … Wikipedia
Orlunda radio transmitter — Orlunda transmitter building today with 86.4 metre tall guyed mast used for mobile phone services … Wikipedia