# High performance liquid chromatography

﻿
High performance liquid chromatography

Infobox chemical analysis
name = High performance liquid chromatography

caption =A HPLC. From left to right: A pumping device generating a gradient of two different solvents, a steel enforced column and an apparatus for measuring the absorbance.
acronym = HPLC
classification =Chromatography
analytes = organic molecules
biomolecules
ions
polymers
manufacturers = Agilent Technologies
Beckman Coulter, Inc.
Hitachi
PerkinElmer, Inc.
Thermo Electron Corporation
Varian, Inc.
Waters Corporation
related = Chromatography
Aqueous Normal Phase Chromatography
Ion exchange chromatography
Size exclusion chromatography
Micellar liquid chromatography
hyphenated = Liquid chromatography-mass spectrometry

High-performance liquid chromatography (or High pressure liquid chromatography, HPLC) is a form of column chromatography used frequently in biochemistry and analytical chemistry to separate, identify, and quantify compounds. HPLC utilizes a column that holds chromatographic packing material (stationary phase), a pump that moves the mobile phase(s) through the column, and a detector that shows the retention times of the molecules. Retention time varies depending on the interactions between the stationary phase, the molecules being analyzed, and the solvent(s) used.

Operation

The sample to be analyzed is introduced in small volume to the stream of mobile phase and is retarded by specific chemical or physical interactions with the stationary phase as it traverses the length of the column. The amount of retardation depends on the nature of the analyte, stationary phase and mobile phase composition. The time at which a specific analyte elutes (comes out of the end of the column) is called the retention time and is considered a reasonably unique identifying characteristic of a given analyte. The use of pressure increases the linear velocity (speed) giving the components less time to diffuse within the column, leading to improved resolution in the resulting chromatogram. Common solvents used include any miscible combinations of water or various organic liquids (the most common are methanol and acetonitrile). Water may contain buffers or salts to assist in the separation of the analyte components, or compounds such as trifluoroacetic acid which acts as an ion pairing agent.

A further refinement to HPLC has been to vary the mobile phase composition during the analysis; this is known as "gradient elution". A normal gradient for reversed phase chromatography might start at 5% methanol and progress linearly to 50% methanol over 25 minutes, depending on how hydrophobic the analyte is. The gradient separates the analyte mixtures as a function of the affinity of the analyte for the current mobile phase composition relative to the stationary phase. This partitioning process is similar to that which occurs during a liquid-liquid extraction but is continuous, not step-wise. In this example, using a water/methanol gradient, the more hydrophobic components will elute (come off the column) under conditions of relatively high methanol (which is hydrophobic); whereas the more hydrophilic compounds will elute under conditions of relatively low methanol/high water. The choice of solvents, additives and gradient depend on the nature of the stationary phase and the analyte. Often a series of tests are performed on the analyte and a number of generic runs may be processed in order to find the HPLC method which gives the best separation of peaks.

Types of HPLC

Normal phase chromatography

Also known as Normal phase HPLC (NP-HPLC), this method separates analytes based on polarity; it was the first kind of HPLC that chemists developed. NP-HPLC uses a polar stationary phase and a non-polar mobile phase, and works effectively for relatively polar analytes. The polar analyte associates with and is retained by the polar stationary phase. Adsorption strengths increase with increased analyte polarity, and the interaction between the polar analyte and the polar stationary phase (relative to the mobile phase) increases the elution time. The interaction strength depends not only on the functional groups in the analyte molecule, but also on steric factors. The effect of sterics on interaction strength allows this method to resolve (separate) structural isomers.

Use of more polar solvents in the mobile phase will decrease the retention time of the analytes, whereas more hydrophobic solvents tend to increase retention times. Very polar solvents in a mixture tend to deactivate the column by occupying the stationary phase surface. This is somewhat particular to normal phase because it is most purely an adsorptive mechanism (the interactions are with a hard surface rather than a soft layer on a surface).

NP-HPLC had fallen out of favor in the 1970s with the development of reversed-phase HPLC because of a lack of reproducibility of retention times as water or protic organic solvents changed the hydration state of the silica or alumina chromatographic media. Recently it has become useful again with the development of HILIC bonded phases which improve reproducibility.

Reversed phase chromatography

Reversed phase HPLC (RP-HPLC or RPC) has a non-polar stationary phase and an aqueous, moderately polar mobile phase. One common stationary phase is a silica which has been treated with RMe2SiCl, where R is a straight chain alkyl group such as C18H37 or C8H17. With these stationary phases, retention time is longer for molecules which are more non-polar, while polar molecules elute more readily. An investigator can also increase retention time by adding a polar solvent to the mobile phase, or decrease retention time by adding a more hydrophobic solvent. RPC is so commonly used that it is often incorrectly referred to as "HPLC" without further specification. The pharmaceutical industry regularly employs RPC to qualify drugs before their release.

RPC operates on the principle of hydrophobic interactions, which result from repulsive forces between a polar eluent, the relatively non-polar analyte, and the non-polar stationary phase. The binding of the analyte to the stationary phase is proportional to the contact surface area around the non-polar segment of the analyte molecule upon association with the ligand in the aqueous eluent. This solvophobic effect is dominated by the force of water for "cavity-reduction" around the analyte and the C18-chain versus the complex of both. The energy released in this process is proportional to the surface tension of the eluent (water: 7.3 × 10-6 J/cm², methanol: 2.2 × 10-6 J/cm²) and to the hydrophobic surface of the analyte and the ligand respectively. The retention can be decreased by adding a less polar solvent (methanol, acetonitrile) into the mobile phase to reduce the surface tension of water. Gradient elution uses this effect by automatically changing the polarity of the mobile phase during the course of the analysis.

Structural properties of the analyte molecule play an important role in its retention characteristics. In general, an analyte with a larger hydrophobic surface area (C-H, C-C, and generally non-polar atomic bonds, such as S-S and others) results in a longer retention time because it increases the molecule's non-polar surface area, which is non-interacting with the water structure. On the other hand, polar groups, such as -OH, -NH2, COO- or -NH3+ reduce retention as they are well integrated into water. Very large molecules, however, can result in an incomplete interaction between the large analyte surface and the ligand's alkyl chains and can have problems entering the pores of the stationary phase.

Retention time increases with hydrophobic (non-polar) surface area. Branched chain compounds elute more rapidly than their corresponding linear isomers because the overall surface area is decreased. Similarly organic compounds with single C-C-bonds elute later than those with a C=C or C-C-triple bond, as the double or triple bond is shorter than a single C-C-bond.

Aside from mobile phase surface tension (organizational strength in eluent structure), other mobile phase modifiers can affect analyte retention. For example, the addition of inorganic salts causes a moderate linear increase in the surface tension of aqueous solutions (ca. 1.5 × 10-7 J/cm² per Mol for NaCl, 2.5 × 10-7 J/cm² per Mol for (NH4)2SO4), and because the entropy of the analyte-solvent interface is controlled by surface tension, the addition of salts tend to increase the retention time. This technique is used for mild separation and recovery of proteins and protection of their biological activity in protein analysis (hydrophobic interaction chromatography, HIC).

Another important component is the influence of the pH since this can change the hydrophobicity of the analyte. For this reason most methods use a buffering agent, such as sodium phosphate, to control the pH. The buffers serve multiple purposes: they control pH, neutralize the charge on any residual exposed silica on the stationary phase and act as ion pairing agents to neutralize charge on the analyte. Ammonium formate is commonly added in mass spectrometry to improve detection of cerain analytes by the formation of ammonium adducts. A volatile organic acid such as acetic acid, or most commonly formic acid, is often added to the mobile phase if mass spectrometry is used to analyze the column eluent. Trifluoroacetic acid is used infrequently in mass spectrometry applications due to its persistence in the detector and solvent delivery system, but can be effective in improving retention of analytes such as carboxylic acids in applications utilizing other detectors, as it is one of the strongest organic acids. The effects of acids and buffers vary by application but generally improve the chromatography.

Reversed phase columns are quite difficult to damage compared with normal silica columns; however, many reversed phase columns consist of alkyl derivatized silica particles and should never be used with aqueous bases as these will destroy the underlying silica particle. They can be used with aqueous acid, but the column should not be exposed to the acid for too long, as it can corrode the metal parts of the HPLC equipment. RP-HPLC columns should be flushed with clean solvent after use to remove residual acids or buffers, and stored in an appropriate composition of solvent. The metal content of HPLC columns must be kept low if the best possible ability to separate substances is to be retained. A good test for the metal content of a column is to inject a sample which is a mixture of 2,2'- and 4,4'- bipyridine. Because the 2,2'-bipy can chelate the metal, the shape of the peak for the 2,2'-bipy will be distorted (tailed) when metal ions are present on the surface of the silica.Fact|date=October 2007..

Size exclusion chromatography

Size exclusion chromatography (SEC), also known as "gel permeation chromatography" or "gel filtration chromatography", separates particles on the basis of size. It is generally a low resolution chromatography and thus it is often reserved for the final, "polishing" step of a purification. It is also useful for determining the tertiary structure and quaternary structure of purified proteins.

This technique is widely used for the molecular weight determination of polysaccharides. SEC is the official technique (suggested by European pharmacopeia) for the molecular weight comparison of different commercially available low-molecular weight heparins.

Ion exchange chromatography

In ion-exchange chromatography, retention is based on the attraction between solute ions and charged sites bound to the stationary phase. Ions of the same charge are excluded. Types of ion exchangers include:
*Polystyrene resins – These allow cross linkage which increases the stability of the chain. Higher cross linkage reduces swerving, which increases the equilibration time and ultimately improves selectivity.
*Cellulose and dextran ion exchangers (gels) – These possess larger pore sizes and low charge densities making them suitable for protein separation.
*Controlled-pore glass or porous silica

In general, ion exchangers favor the binding of ions of higher charge and smaller radius.

An increase in counter ion (with respect to the functional groups in resins) concentration reduces the retention time. An increase in pH reduces the retention time in cation exchange while a decrease in pH reduces the retention time in anion exchange.

This form of chromatography is widely used in the following applications: water purification, preconcentration of trace components, ligand-exchange chromatography, ion-exchange chromatography of proteins, high-pH anion-exchange chromatography of carbohydrates and oligosaccharides, and others.

Bioaffinity chromatography

This chromatographic process relies on the property of biologically active substances to form stable, specific, and reversible complexes. The formation of these complexes involves the participation of common molecular forces such as the Van der Waals interaction, electrostatic interaction, dipole-dipole interaction, hydrophobic interaction, and the hydrogen bond. An efficient, biospecific bond is formed by a simultaneous and concerted action of several of these forces in the complementary binding sites.

A composition of the mobile phase that remains constant throughout the procedure is termed "isocratic".

In contrast to this is the so called "gradient elution", which is a separation where the mobile phase changes its composition during a separation process. One example is a gradient in 20 min starting from 10% methanol and ending up with 30% methanol. Such a gradient can be increasing or decreasing. The benefit of gradient elution is that it helps speed up elution by allowing components that elute more quickly to come off the column under different conditions than components which are more readily retained by the column. By changing the composition of the solvent, components that are to be resolved can be selectively more or less associated with the mobile phase. As a result, at equilibrium they spend more time in the solvent and less time in the stationary phase, and therefore they elute faster.

Parameters

Internal diameter

The internal diameter (ID) of an HPLC column is a critical aspect that influences sensitivity and determines the quantity of analyte that can be loaded onto the column. Larger columns are usually seen in industrial applications such as the purification of a drug product for later use. Low ID columns have improved sensitivity and lower solvent consumption at the expense of loading capacity.
*Larger ID columns (over 10 mm) are used to purify usable amounts of material because of their large loading capacity.
*Analytical scale columns (4.6 mm) have been the most common type of columns, though smaller columns are rapidly gaining in popularity. They are used in traditional quantitative analysis of samples and often use a UV-Vis absorbance detector.
*Narrow-bore columns (1-2 mm) are used for applications when more sensitivity is desired either with special UV-vis detectors, fluorescence detection or with other detection methods like liquid chromatography-mass spectrometry
*Capillary columns (under 0.3 mm) are used almost exclusively with alternative detection means such as mass spectrometry. They are usually made from fused silica capillaries, rather than the stainless steel tubing that larger columns employ.

Particle size

Most traditional HPLC is performed with the stationary phase attached to the outside of small spherical silica particles (very small beads). These particles come in a variety of sizes with 5 $mu$m beads being the most common. Smaller particles generally provide more surface area and better separations, but the pressure required for optimum linear velocity increases by the inverse of the particle diameter squared. [ [http://www.lcgceurope.com/lcgceurope/article/articleDetail.jsp?id=333246&pageID=3# Fast and Ultrafast HPLC on sub-2 μm Porous Particles — Where Do We Go From Here? - LC-GC Europe ] ] [ cite journal | last = Xiang | first = Y. | coauthors = Liu Y. and Lee M.L. | year = 2006 |date=2006 | title = Ultrahigh pressure liquid chromatography using elevated temperature| journal = Journal of Chromatography A | volume = 1104 | issue = 1-2 | pages = 198–202 | doi = 10.1016/j.chroma.2005.11.118 ] [ cite journal | last = Horváth | first = Cs. | coauthors = Preiss B.A. and Lipsky S.R. | year = 1967 |date=1967 | title = | journal = Analytical Chemistry | volume = 39 | pages = 1422–1428 | doi = 10.1021/ac60256a003 ]

This means that changing to particles that are half as big, keeping the size of the column the same, will double the performance, but increase the required pressure by a factor of four. Larger particles are used in preparative HPLC (column diameters 5 cm up to >30 cm) and for non-HPLC applications such as solid-phase extraction.

Pore size

Many stationary phases are porous to provide greater surface area. Small pores provide greater surface area while larger pore size has better kinetics, especially for larger analytes. For example, a protein which is only slightly smaller than a pore might enter the pore but not easily leave once inside.

Pump pressure

Pumps vary in pressure capacity, but their performance is measured on their ability to yield a consistent and reproducible flow rate. Pressure may reach as high as 40 MPa (6000 lbf/in2), or about 400 atmospheres. Modern HPLC systems have been improved to work at much higher pressures, and therefore are able to use much smaller particle sizes in the columns (<2 $mu$m). These "Ultra High Performance Liquid Chromatography" systems or UHPLCs can work at up to 100 MPa (15,000 lbf/in²), or about 1000 atmospheres. The term "UPLC", though sometimes used is a trademark of Waters Corporation and not the name for the technique in general.

Manufacturers of HPLC chromatographs

*Anachem
*Agilent Technologies
*Beckman Coulter, Inc.
*Hitachi
*PerkinElmer, Inc.
*Thermo Fisher Scientific
*Varian, Inc.
*Waters Corporation

Manufacturers of HPLC columns and accessories

*Agilent Technologies
*AkzoNobel
*Beckman Coulter, Inc.
*Merck KGaA
*Phenomenex
*SGE Analytical Science
*Sigma-Aldrich
*Thermo Fisher Scientific
*Tosoh Corporation
*Varian, Inc.
*Waters Corporation
*W. R. Grace and Company

*Ion chromatography

References

* [http://www.waters.com/WatersDivision/ContentD.asp?watersit=JDRS-6VRHBW HPLC Primer] - A detailed primer with history, glossary, and technology sections
* [http://www.mtc-usa.com/calculators_chrom.asp Scientific Calculator for HPLC]
* [http://www.iest.org/i4a/pages/index.cfm?pageid=3496 IEST]

Wikimedia Foundation. 2010.

### Look at other dictionaries:

• High-performance liquid chromatography — An HPLC. From left to right: A pumping device generating a gradient of two different solvents, a steel enforced column and an apparatus for measuring the absorbance. Acronym HPLC Classification Chromatography …   Wikipedia

• High Performance Liquid Chromatography — Hochleistungsflüssigkeitschromatographie (engl. high performance liquid chromatography, HPLC) in den Anfangszeiten dieser Technik auch Hochdruckflüssigchromatographie (engl. high pressure liquid chromatography) genannt ist eine analytische… …   Deutsch Wikipedia

• high performance liquid chromatography — efektyvioji skysčių chromatografija statusas T sritis chemija apibrėžtis Skysčių chromatografijos metodas, kai judančioji fazė spaudžiama per chromatografinę kolonėlę dideliu slėgiu. atitikmenys: angl. high performance liquid chromatography; high …   Chemijos terminų aiškinamasis žodynas

• high-performance liquid chromatography — high per·for·mance liquid chromatography pə(r) fȯr mən(t)s n liquid chromatography in which the degree of separation is increased by forcing a solvent under pressure through a densely packed adsorbent abbr. HPLC called also high pressure liquid… …   Medical dictionary

• high-performance liquid chromatography — noun : liquid chromatography in which the degree of separation is increased by forcing a solvent under pressure through a densely packed adsorbent abbreviation HPLC; called also high pressure liquid chromatography …   Useful english dictionary

• high-performance liquid chromatography — noun high pressure liquid chromatography …   Wiktionary

• Denaturing High Performance Liquid Chromatography — La Denaturing High Performance Liquid Chromatography (DHPLC) est une méthode chromatographique permettant la détection de substitutions de bases, de petites délétions ou d insertions au niveau de l ADN. Grâce à sa rapidité et sa résolution… …   Wikipédia en Français

• high-pressure liquid chromatography — high pres·sure liquid chromatography presh ər n high performance liquid chromatography …   Medical dictionary

• high performance power chromatography — efektyvioji skysčių chromatografija statusas T sritis chemija apibrėžtis Skysčių chromatografijos metodas, kai judančioji fazė spaudžiama per chromatografinę kolonėlę dideliu slėgiu. atitikmenys: angl. high performance liquid chromatography; high …   Chemijos terminų aiškinamasis žodynas

• high pressure liquid chromatography — efektyvioji skysčių chromatografija statusas T sritis chemija apibrėžtis Skysčių chromatografijos metodas, kai judančioji fazė spaudžiama per chromatografinę kolonėlę dideliu slėgiu. atitikmenys: angl. high performance liquid chromatography; high …   Chemijos terminų aiškinamasis žodynas