Electrochemical gas sensor
Electrochemical gas sensors are
gas detectorsthat measure the volume of a target gasby oxidizing or reducing the target gas at an electrodeand measuring the resulting current.
The sensors contain two or three electrodes, occasionally four, in contact with an
electrolyte. The electrodes are typically fabricated by fixing a high surface area precious metal on to the porous hydrophobic membrane. The working electrode contacts both the electrolyte and the ambient air to be monitored usually via a porous membrane. The electrolyte most commonly used is a mineral acid, but organic electrolytes are also used for some sensors. The electrodes and housing are usually in a plastic housing which contains a gas entry hole for the gas and electrical contacts.
Theory of operation
The gas diffuses into the sensor, through the back of the porous membrane to the working electrode where it is oxidized or reduced. This electrochemical reaction results in an electric current that passes through the external circuit. In addition to measuring, amplifying and performing other signal processing functions, the external circuit maintains the voltage across the sensor between the working and counter electrodes for a two electrode sensor or between the working and reference electrodes for a three electrode cell. At the counter electrode an equal and opposite reaction occurs, such that if the working electrode is an oxidation, then the counter electrode is a reduction.
Diffusion controlled response
The magnitude of the current is controlled by how much of the target gas is oxidized at the working electrode. Sensors are usually designed so that the gas supply is limited by diffusion and thus the output from the sensor is
linearly proportional to the gas concentration. This linear output is one of the advantages of electrochemical sensors over other sensor technologies, (e.g. infrared), whose output must be linearized before they can be used. A linear output allows for more precise measurement of low concentrations and much simpler calibration (only baseline and one point are needed).
Diffusion control offers another advantage. Changing the diffusion barrier allows the sensor manufacturer to tailor the sensor to a particular target gas concentration range. In addition, since the diffusion barrier is primarily mechanical, the calibration of electrochemical sensors tends to be more stable over time and so electrochemical sensor based instruments require much less maintenance than some other detection technologies. In principle, the sensitivity can be calculated based on the diffusion properties of the gas path into the sensor, though experimental errors in the measurement of the diffusion properties make the calculation less accurate than calibrating with test gas. [“Amperometric Gas Sensor Response Times,” P.R. Warburton, M.P. Pagano, R. Hoover, M. Logman, and K. Crytzer, Yi.J. Warburton; Anal. Chem., 70 (5), 998 -1006, 1998.]
For some gases such as
ethylene oxide, cross sensitivity can be a problem because ethylene oxide requires a very active working electrode catalyst and high operating potential for its oxidation. Therefore gases which are more easily oxidized such as alcohols and carbon monoxidewill also give a response. Cross sensitivity problems can be eliminated though through the use of a chemical filter, for example filters that allows the target gas to pass through unimpeded, but which reacts with and removes common interferences.
While electrochemical sensors offer many advantages, they are not suitable for every gas. Since the detection mechanism involves the oxidation or reduction of the gas, electrochemical sensors are usually only suitable for gases which are electrochemically active, though it is possible to detect electrochemically inert gases indirectly if the gas interacts with another species in the sensor that then produces a response. [D. Pletcher, J. Evans, P.R. Warburton, T.K. Gibbs, US Patent 5,071,526, Dec. 10, 1991, “Acidic Gas Sensors and Method of Using the Same”] Sensors for carbon dioxide are an example of this approach and they have been commercially available for several years.
Carbon monoxide detector
Wikimedia Foundation. 2010.
Look at other dictionaries:
Gas leak detection — is the detection of gas leaks with apparatus like catalytic/infrared point sensors and IR open path sensors, more recent detection methods include Ultrasonic Gas Leak Detectors.In order to detect gas leaks with traditional methods, the gas itself … Wikipedia
Gas detector — A gas detector is a device which detects the presence of various gases within an area, usually as part of a system to warn about gases which might be harmful to humans or animals. Gas detectors can be used to detect combustible, toxic, and oxygen … Wikipedia
Oxygen sensor — Contents 1 Automotive applications 1.1 Function of a lambda probe 1.2 The probe … Wikipedia
Potentiometric sensor — A potentiometric sensor is a type of chemical sensor that may be used to determine the analytical concentration of some components of the analyte gas or solution. These sensors measure the electrical potential of an electrode when no current is… … Wikipedia
Glossary of fuel cell terms — The Glossary of fuel cell terms lists the definitions of many terms used within the fuel cell industry. The terms in this glossary may be used by fuel cell industry associations, in education material and fuel cell codes and standards to name but … Wikipedia
Carbon monoxide poisoning — This article is about carbon monoxide poisoning. For general information on carbon monoxide, see carbon monoxide. Carbon monoxide poisoning Classification and external resources Spacefilling model of a carbon monoxide molecule … Wikipedia
Carbon monoxide breath monitor — A carbon monoxide breath monitor (breath CO monitor) is a device used to measure the level of carbon monoxide in a user s exhalation. The device is used both as a motivational and educational tool for smoking cessation and also as a clinical aid… … Wikipedia
Quartz crystal microbalance — A quartz crystal microbalance (QCM) measures a mass per unit area by measuring the change in frequency of a quartz crystal resonator. The resonance is disturbed by the addition or removal of a small mass due to oxide growth/decay or film… … Wikipedia
Carbon monoxide detector — connected to a North American power outlet A carbon monoxide detector or CO detector is a device that detects the presence of the carbon monoxide (CO) gas in order to prevent carbon monoxide poisoning. CO is a colorless and odorless compound… … Wikipedia
Microelectromechanical systems — (MEMS) (also written as micro electro mechanical, MicroElectroMechanical or microelectronic and microelectromechanical systems) is the technology of very small mechanical devices driven by electricity; it merges at the nano scale into… … Wikipedia