# Absorption wavemeter

﻿
Absorption wavemeter

The Absorption wavemeter is a simple device for the measurement of radio frequency energy at different frequencies.

The device can be used by the users of radio equipment to check the approximate frequency of a strong signal source, and also to check the output of a transmitter for harmonics. Many radio amateurs use one to check the purity and frequency of their transmitters [http://www.darleys.pwp.blueyonder.co.uk/Radio/advanced/measure1.htm] [http://www.tpub.com/content/neets/14188/css/14188_63.htm] [http://www.brightbell.com/radio/antenna1.html#awavemeter] . Also devices can be made which work in the same way that are able to detect mobile phones [http://www.creative-science.org.uk/absorb.html] . As an alternative a dip meter can be used.

HF and VHF

The most simple form of the device is a variable capacitor with a coil wired across its terminals. Attached to one the terminals of the LC circuit is a diode , then between the end of the diode not wired to the LC circuit and the terminal of the LC circuit not bearing the diode is wired a ceramic decoupling capacitor. Finally a galvanometer is wired to the terminals of the decoupling capacitor.

The device will be sensitive to strong sources of radiowaves at the frequency at which the LC circuit is resonant.

This is given by f = 1/2π L0.5C0.5

When the device is exposed to an RF field which is at the resonant frequency a DC voltage will appear on the terminals on the left hand side. The coil is often outside the case of the unit so it can be brought close to the object being probed.

= UHF and SHF =

At the higher frequencies it is not possible to use lumped components for the tuned circuit. Instead methods such as stripline are used. One design for UHF and SHF is a resonant λ/4 (quarter wave) rod which can vary in length. Another design for X-band (10 GHz) is a resonator cavity which can be changed in length.

As an alternative for UHF, Lecher transmission lines can be used [http://www.intuitor.com/resonance/standingEwaveDemo.html] . It is possible to measure roughtly the frequency of a transmitter using Lecher lines.

* Grid dip oscillator

Wikimedia Foundation. 2010.

### Look at other dictionaries:

• absorption wavemeter — sugertinis bangomatis statusas T sritis Standartizacija ir metrologija apibrėžtis Bangomatis, kurio veikimas pagrįstas elektromagnetinės energijos sugertimi. atitikmenys: angl. absorption wavemeter vok. Absorptionswellenmesser, m rus.… …   Penkiakalbis aiškinamasis metrologijos terminų žodynas

• absorption wavemeter — sugertinis bangomatis statusas T sritis fizika atitikmenys: angl. absorption wavemeter vok. Absorptionswellenmesser, m rus. абсорбционный волномер, m pranc. ondemètre à absorption, m …   Fizikos terminų žodynas

• absorption wavemeter — A wavemeter which depends on a resonance absorption in a tuned circuit, constructed with very stable inductance and capacitance …   Dictionary of automotive terms

• wavemeter — /wayv mee teuhr/, n. a device for measuring the wavelength or frequency of a radio wave. [1900 05; WAVE + METER] * * * ▪ measurement device       device for determining the distance between successive wavefronts of equal phase along an… …   Universalium

• ondemètre à absorption — sugertinis bangomatis statusas T sritis Standartizacija ir metrologija apibrėžtis Bangomatis, kurio veikimas pagrįstas elektromagnetinės energijos sugertimi. atitikmenys: angl. absorption wavemeter vok. Absorptionswellenmesser, m rus.… …   Penkiakalbis aiškinamasis metrologijos terminų žodynas

• ondemètre à absorption — sugertinis bangomatis statusas T sritis fizika atitikmenys: angl. absorption wavemeter vok. Absorptionswellenmesser, m rus. абсорбционный волномер, m pranc. ondemètre à absorption, m …   Fizikos terminų žodynas

• Microwave — This article is about the electromagnetic wave. For the cooking appliance, see Microwave oven. For other uses, see Microwaves (disambiguation). A microwave telecommunications tower on Wrights Hill in Wellington, New Zealand Microwaves, a subset… …   Wikipedia

• Grid dip oscillator — Dipmeter could also refer to an influential early commercial expert system called Dipmeter Advisor; or may refer to an instrument that measures the dip angle of the planet s magnetic field, the field line angle in a vertical plane. A dipmeter and …   Wikipedia

• Absorptionswellenmesser — sugertinis bangomatis statusas T sritis Standartizacija ir metrologija apibrėžtis Bangomatis, kurio veikimas pagrįstas elektromagnetinės energijos sugertimi. atitikmenys: angl. absorption wavemeter vok. Absorptionswellenmesser, m rus.… …   Penkiakalbis aiškinamasis metrologijos terminų žodynas

• sugertinis bangomatis — statusas T sritis Standartizacija ir metrologija apibrėžtis Bangomatis, kurio veikimas pagrįstas elektromagnetinės energijos sugertimi. atitikmenys: angl. absorption wavemeter vok. Absorptionswellenmesser, m rus. абсорбционный волномер, m pranc.… …   Penkiakalbis aiškinamasis metrologijos terminų žodynas